 Research
 Open Access
 Published:
Compact orthogonaldispersion device using a prism and a transmission grating
Journal of the European Optical SocietyRapid Publications volume 14, Article number: 8 (2018)
Abstract
Background
Spectroscopy was one of the main applications of gratings. Most spectrometers today are based on reflection grating designs, since high quality transmission grating technologies have only matured over the last several years. Conventional crossdispersion systems are based on the reflection grating. For traditional directvision or constantdispersion grism, both the prism and grating have the same dispersion direction.
Methods
This paper presents an orthogonaldispersion device based on a transmission grating attached to a prism. The dispersion directions of the prism and transmission grating are orthogonal. The analytical expressions for the dispersion of this device are derived in detail. The numerical results and raytracing simulations by ZEMAX software are shown.
Results
The simulation results show that the spectrum is spread over 5 diffraction orders, m=3 to 1, the nondiffracted order m=0 is straight, and the other orders are curved.
Conclusions
The device will be able to provide a compact, smallsized and broadband orthogonal dispersion device that is suitable for the medium resolution spectrometer.
Background
Spectroscopy was one of the earliest applications of gratings, and it remains one of the main grating uses [1]. More particularly, the echelle grating suggested first by Harrison [2] is preferred in a number of spectrometers. The echelle grating is usually used in high diffraction orders that will overlap. To solve this problem, traditional echelle spectrometers usually employ cutoff filters or a cross disperser in the form of lowdispersion gratings or prisms. An early echelle spectrometer uses two planar reflection gratings, i.e., an echelle grating and a reflection grating, operating in two perpendicular planes [3]. The resulting dispersion produces a twodimensional spectrogram including several lines side by side that correspond to different diffraction orders, namely, several diffractionorder lines side by side with the wavelength varying with its position along each line. As an improved echelle spectrometer [4], the second grating is replaced by a combination of a reflection grating with a prism so that the light from the echelle grating is refracted into the prism, diffracted by the grating, then refracted back out of the prism. The diffraction orders can be spread out almost evenly by choosing the prism parameters, allowing more of the detector area to record useful information. For the above referenced echelle spectrometers, both the incident and diffracted light rays are perpendicular to the grating grooves.
The diffraction grating can also be used in the conical diffraction mode [5,6,7,8,9,10] in which the incident light beam is not perpendicular to the grating grooves, thus the diffracted light rays lie on a circular cone rather than in a plane. Typically, the conical diffraction spectrometers reported in [11] uses a Littrow configuration coupled with a shared focusing lens and a linear detector array, and they are compact with a substantially reduced size. Nevertheless, these two designs are unable to separate the overlapping diffraction orders of the echelle grating. On the contrary, the Littrow conical diffraction spectrometer proposed by [12] includes an additional prism for separating the overlapping diffraction orders of the echelle grating to produce twodimensional high dispersion without requiring a large field of view.
Most spectrometers today are based on reflection grating designs [13,14,15], since high quality transmission grating technologies have only matured over the last several years. However, it is currently possible to produce high quality transmission gratings that rival reflection gratings in all aspects. Accordingly, transmission gratings can be used as dispersive devices in the infrared, visible or nearultraviolet light [16,17,18,19,20,21]. Transmission gratings offer low alignment sensitivity, which helps minimize alignment errors. A transmission grating offers a basic simplicity for optical designs that can be beneficial in spectral separation. In certain types of instruments, transmission gratings are much more convenient to use than reflection gratings. For example, a configuration inserts a transmission grating in front of the lens, which can be used for studying the composition of falling meteors or the reentry of space vehicles, where the distant luminous streak becomes the entrance slit.
The directvision grism combines a lowdispersion transmission grating with a beambending prism to provide inline viewing for a single wavelength [22]. The purpose of the prism is simply to change the direction of a given diffracted wavelength, for zero net deviation. The constantdispersion grism consists of a prism and a transmission grating to disperse the spectrum so that the spectrum fringes are equally spaced across the detector. Namely, for the constantdispersion grism the angular dispersion with respect to wave number is approximately constant. Both the transmission grating and prism contribute to the dispersion, and the net deviation angles are not constrained to be zero. For the directvision or constantdispersion grism, both the prism and transmission grating have the same dispersion direction.
This paper is a theoretical study of a compact, smallsized and broadband orthogonaldispersion prismgrating device used for spectrometers. Section 2 presents the basic principle and derives the analytical expressions for the dispersion of the device. Section 3 shows the numerical results and raytracing simulations by ZEMAX software. The last section gives the conclusion.
Methods
Figure 1a shows the optical layout of the orthogonaldispersion prismgrating device (PGD) that combines a prism with a transmission grating. The dispersion directions of the prism and transmission grating are orthogonal. The transmission grating is superimposed on the hypotenuse surface of an optical wedge, and the rightangle surface of the optical wedge is directly attached to one surface of the prism. The grating normal is parallel to yaxis, the grating grooves are parallel to zaxis, and the y´axis is perpendicular to the rightangle surface of the optical wedge. The prism disperses the incident light along the vertical direction (i.e., zaxis), then the transmission grating disperses the separated beams along the horizontal direction (i.e., xaxis). The spectrum is spread out in two dimensions.
Suppose that γ denotes the vertex angle of the prism, γ_{1} denotes the vertex angle of the optical wedge, n_{ p } (λ_{ i }) is the refractive index of the prism for wavelength λ_{ i }, and n_{ g } (λ_{ i }) is the refractive index of the grating material for wavelength λ_{ i }.
Figure 1b shows the ray tracing in the sagittal plane (i.e., x´y´ plane), in which δ is the angle between the inward normal vector of the grating and the projection of the incident wave vector \( \overrightarrow{k} \) onto the normal plane of the grating (i.e., xy plane). It is easily obtained that δ=γ_{1}.
Figure 1c depicts the ray tracing in the meridian plane (i.e., y´z´ plane), in which ε is the angle between the incident wave vector \( \overrightarrow{k} \) and the normal plane of the grating. A ray enters the prism at angle θ, is refracted at angle θ_{1}, leaves the prism at angle θ_{2} , and is refracted into the grating material at angle θ_{3}. It is easily obtained that
From Eq. (1ad), the angle ε is given by
Let the plane of incidence be the plane that is made up of the incident light ray and the grating normal. Let the normal plane of the grating be the plane that is perpendicular to the grating grooves. In the conical diffraction case, the grating equation of a plane transmission grating is given by [1].
where m is the diffraction order, λ_{ i } is the wavelength of light, d is the groove spacing of grating, ε is the angle between the incident light path and the plane perpendicular to the grating grooves, α is the angle of incidence measured from the grating normal, β_{ m } is the diffraction angle measured from the grating normal. The angle sign convention is that angles measured counterclockwise from the normal are positive and angles measured clockwise from the normal are negative.
Figure 2(a) shows the conical diffraction by a plane transmission grating, in which the grating lies in the xz plane, the grating grooves are parallel to zaxis, and the grating normal is parallel to yaxis. Both the incident ray and the reflected rays lie in the y < 0 halfspace; the transmitted diffracted rays lie in the y > 0 halfspaceThat is, the incident light is dispersed on the opposite side of the transmission grating. All diffracted rays fall on the surface of a halfcone whose cone axis is parallel to the grating grooves.
Figure 2(b) shows the geometry diagram of a diffracted light ray. The direction of the diffracted ray with a wave vector \( {\overrightarrow{k}}_m={k}_{mx}{\overrightarrow{e}}_x+{k}_{my}{\overrightarrow{e}}_y+{k}_{mz}{\overrightarrow{e}}_z \) is given by two parameters: (1) φ_{ m } is the angle between the diffracted ray and the normal plane of the grating and (2) ρ_{ m } is the angle between the outward normal vector of the grating and the projection of the diffracted ray onto the normal plane of the grating. We can get
Based on Eqs. (4ad), it can be obtained that
Figure 2(c) shows the geometry diagram of an incident wave vector. A light ray, with a wave vector \( \overrightarrow{k}={k}_x{\overrightarrow{e}}_x+{k}_y{\overrightarrow{e}}_y+{k}_z{\overrightarrow{e}}_z \) and its modulus k = 2π/λ, is incident on the grating at an offplane direction (ε ≠ 0). We can obtain
From Eqs. (6ad), it can be obtained that
According to the law of refraction and the geometry, we can get
Based on Eqs. (2) and (8), the elevation angle φ_{ m }, for the PGD, is given by
Based on Eqs. (3), (5), (7) and (8), the azimuth angle ρ_{ m }, for the PGD, is given by
Eqs. (9) and (10) completely specify the diffracted ray angles in terms of the incident ray angles. For an incident ray of a given wavelength, the diffracted rays will all have the same value of φ_{ m } given in Eq. (9) and will therefore all lie in a common conical surface, as shown in Fig. 2(a). For reference, if ε = 0, the conical diffraction grating equation [Eq. (3)] simplifies to yield the classical inplane diffraction grating equation, mλ_{ i } = d[n_{ g }(λ_{ i }) sin α − sin β_{ m }], for a plane transmission grating.
Results and discussion
For a source spectrum with a spectral range from 500 nm to 800 nm, for example, we can consider a transmission grating with 75 grooves/mm and a prism that is made of Zinc sulfide (ZnS) with a refractive index formula as [23].
Suppose that the vertex angle of the prism is γ = 20^{°}, the angle of incidence on the prism is θ = 20^{°}, and the vertex angle of the optical wedge is γ_{1} = 12^{°}. According to Eqs. (9)–(11), the separation of the spectral orders is illustrated in Fig. 3, in which the spectrum is spread over 5 diffraction orders, m = 3 to − 1, with the azimuth angle from small to big between 15 and 65 degrees. Moreover, the nondiffracted order m = 0 is straight, but the other orders are curved.
Suppose that the entrance pupil diameter is 10 mm, the center thickness of the prism is 22 mm, the center thickness of the grating is 15 mm. For a wavelength range from 500 nm to 900 nm in diffraction order − 1 through 3, the raytracing simulations by ZEMAX software for the orthogonaldispersion PGD are shown in the following several figures. Figure 4 shows one Shaded Model, Fig. 5 shows the other Shaded Model in which the different color rays represent the different diffraction orders, Fig. 6 shows one Spot Diagram in which the different color rays represent the different diffraction orders, and Fig. 7 shows another Spot Diagram in which the different color rays represent the different wavelengths. According to Figs. 6 and 7, the nondiffracted order m = 0 is straight, but the other orders are curved especially in the − 1 order.
Conclusions
An orthogonaldispersion PGD was investigated in theory. The raytracing simulations by ZEMAX indicated the accuracy of the theoretical analysis and mathematical formulas. In order to improve the spectral resolution, the dispersive power of the PGD needs to be improved, which can be realized by either appropriately increasing the vertex angle of the prism, properly reducing the groove spacing of grating, or both. In addition, the dispersive power of the PGD can also be changed by choosing different materials of the prism and grating. There are two main features to the PGD. First, the orthogonaldispersion PGD is based on the transmission grating, whereas traditional crossdispersion systems are based on the reflection grating. Second, the transmission grating is superimposed on the hypotenuse surface of an optical wedge with a rightangle surface directly attached to the prism, which not only makes the PGD compact, but also makes the airglass interfaces fewer and therefore makes the reflection loss less. The unique design makes the PGD capable of providing a compact, smallsized and broadband orthogonaldispersion device that is applicable to the medium resolution spectrometers.
Abbreviations
 PGD:

Prismgrating device
References
Palmer, C., Loewen, E.: Diffraction Grating Handbook. Newport Corporation, New York (2005)
Harrison, G.R.: The production of diffraction gratings: II. The design of echelle gratings and spectrographs. J. Opt. Soc. Am. 39(7), 522–527 (1949)
Dantzler, A.A.: Echelle spectrograph software design aid. Appl. Opt. 24(24), 4504–4508 (1985)
Dantzler, A.A.: Echelle grism spectrograph. Appl. Opt. 25(23), 4267–4287 (1986)
Cash, W.: Echelle spectrographs at grazing incidence. Appl. Opt. 21(4), 710–717 (1982)
Cornet, P., Chandezon, J., Faure, C.: Conical diffraction of a plane wave by an inclined parallelplate grating. J. Opt. Soc. Am. A. 14(2), 437–449 (1997)
Goray, L.I., Schmidt, G.: Solving conical diffraction grating problems with integral equations. J. Opt. Soc. Am. A. 27(3), 585–597 (2010)
Werner, W.: Xray and extended UV spectrometer designs based on offplane grating mountings. Space Sci. Rev. 29(4), 455–459 (1981)
Werner, W.: Comparison of various grazing incidence spectrometer designs based on conical diffraction. Appl. Opt. 23(14), 2408–2411 (1984)
Lemaire, P.: Ultraviolet conical diffraction: a nearstigmatic tandem grating mounting spectrometer. Appl. Opt. 30(10), 1294–1302 (1991)
Everett, M.J., Zhou, Y., Horn, J.M.M., O’Hara, K., Foley, J.P.: Littrow spectrometer and a spectral domain optical coherence tomography system with a Littrow spectrometer. US patent 7456957 B2, United States (2008)
Yang, Q.: Compact highresolution Littrow conical diffraction spectrometer. Appl. Opt. 55(18), 4801–4807 (2016)
Ballester, P., Rosa, M.R.: Modeling echelle spectrographs. Astron. Astrophys. Suppl. Ser. 126(3), 563–571 (1997)
Ge, J., Angel, J.R.P., Jacobsen, B., Woolf, N., Fugate, R.Q., Black, J.H., LloydHart, M.: An optical ultrahighresolution crossdispersed echelle spectrograph with adaptive optics. Publ. Astron. Soc. Pac. 114(798), 879–891 (2002)
Yin, L., Bayanheshig, Yang, J., Lu, Y., Zhang, R., Sun, C., Cui, J.: Highaccuracy spectral reduction algorithm for the échelle spectrometer. Appl. Opt. 55(13), 3574–3581 (2016)
Yang, J., Ding, Y., Zhang, W., Zhang, J., Zheng, Z.: Precise measurement technology of softxray spectrum using dual transmission grating spectrometer. Rev. Sci. Instrum. 74(10), 4268–4272 (2003)
Weisberg, A., Craparo, J., De Saro, R., Pawluczyk, R.: Comparison of a transmission grating spectrometer to a reflective grating spectrometer for standoff laserinduced breakdown spectroscopy measurements. Appl. Opt. 49(13), C200–C210 (2010)
Wang, C., Ding, Z., Mei, S., Yu, H., Hong, W., Yan, Y., Shen, W.: Ultralongrange phase imaging with orthogonal dispersive spectraldomain optical coherence tomography. Opt. Lett. 37(21), 4555–4557 (2012)
Bao, W., Ding, Z., Li, P., Chen, Z., Shen, Y., Wang, C.: Orthogonal dispersive spectraldomain optical coherence tomography. Opt. Express. 22(8), 10081–10090 (2014)
Xu, K.: Integrated silicon directly modulated light source using pwell in standard CMOS technology. IEEE Sensors J. 16(16), 6184–6191 (2016)
Xu, K., Li, G.P.: A novel way to improve the quantum efficiency of silicon lightemitting diode in a standard silicon complementary metal–oxide–semiconductor technology. J. Appl. Phys. 113(10), 103106 (2013)
Schroeder, D.J.: Astronomical optics. Academic, San Diego (1987)
Bass, M., Li, G., Stryland, E.V.: Handbook of Optics. McGrawHill, New York (2010)
Acknowledgements
None.
Funding
This work was supported by the National Natural Science Foundation of China (NSFC) (Grant number 61605151).
Availability of data and materials
The data and materials supporting the conclusions of this article is included within the article.
Author information
Authors and Affiliations
Contributions
QY has come up with the idea, derived the formulas, and performed the raytracing simulations by ZEMAX software. QY was the First Author and Corresponding Author. WW performed numerical results. Both authors read and approved the final manuscript.
Corresponding author
Ethics declarations
Competing interests
The authors declare that they have no competing interests.
Publisher’s Note
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Rights and permissions
Open Access This article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made.
About this article
Cite this article
Yang, Q., Wang, W. Compact orthogonaldispersion device using a prism and a transmission grating. J. Eur. Opt. Soc.Rapid Publ. 14, 8 (2018). https://doi.org/10.1186/s4147601800779
Received:
Accepted:
Published:
DOI: https://doi.org/10.1186/s4147601800779
Keywords
 Orthogonal dispersion
 Transmission grating
 Prism
 Spectrometer